Task 1: Design, Fabrication, and Testing Tissue Equivalent Proportional Counters (TEPC) Detectors The purpose of this task was to design, build, and assemble a prototype Tissue Equivalent Proportional Counters (TEPC) that would satisfy the basic specifications outlined by NASA for a dosimeter for astronauts during lunar EVAs (extravehicular activities) and as area monitors in space craft and habitats. The spherical TEPC is based on a single-wire anode with recessed guard ring insulators to shape the electric field near the poles. The diameter of the gas cavity is 18 mm and the wall thickness is 3 mm for a total diameter of 24 mm (~1 inch). Aluminum vacuum chambers with a shell thickness of 0.5 mm were designed and gold plated to maintain electrical conductivity. A system using a high sensitivity mass spectrometer was assembled to measure vacuum leaks for the assembled detectors with high special specificity. We have been using a version of the software package LORENTZ 3D to model the electric field inside a spherical detector with a linear collector. This uses special modeling techniques based on the Boundary Element Method to make the solution of these very challenging problems a simple matter. The geometry of the problem can be created with the geometric modeler built into the electric field solvers or can be imported from any of the major CAD (computer-aided design) vendors. More importantly, the geometry can be changed parametrically to optimize a design for robustness, weight, size and, of course, cost. We have fabricated seven versions of the TEPC and Vacuum Chamber. Three versions with spherical detector and single wire anode operated with the wall at high voltage and the anode at ground were delivered for electronics development and testing. Two versions with a spherical detector and single wire anode operated with the wall at ground and the anode at high voltage have been used for TEPC development and comparative analysis. Two versions of a detector using a new hybrid design with a parallel wire grid surrounding the anode we designed and built. The objective of this design was to form a virtual cylindrical geometry around the anode with that would improve the spatial resolution of the TEPC without distorting the signals required for microdosimetry applications. The single and multi wire detectors with grounded anodes were exposed to Neutrons from a PuBe source at Colorado State University and high energy charged particle beams at the Heavy Ion Medical Accelerator in Chiba (HIMAC) synchrotron in Chiba Japan and the NASA Space Radiation Laboratory (NSRL) at Brookhaven National Laboratory. This included the following ions and energies: 56Fe (380 MeV/amu), 18Ar (300 MeV/amu), 12C (200 MeV/amu), and 1H (230 MeV/amu). Measurements were taken at several angles of incidence to determine the angular response of the detector. These results were compared with similar measurements using a commercial TEPC with a Rossi design that has a helical grid surrounding the anode to provide a uniform angular response. Analysis of detector response using digital signal processing was initiated. This system replaced a preamplifier and shaping amplifier with a programmable logic device (PLD) that captures the output signal from the TEPC and digitizes the amplitude in 10 nsec intervals. Task 2: Modeling Detector Response The objective of this task is to determine the response of the TEPC under ambient conditions and during solar particle events (SPE) on the lunar surface. Computations using the Monte Carlo Code PHITS have been made to determine the energy deposition in the TEPC using protons with an energy spectrum from a SPE in October 2003. These data were compared with the dose that would be delivered to the skin beneath a space suit with an areal density of 0.4 g/cm2. It is clear that a stainless steel vacuum chamber in Mod 1 needs to be replaced with lighter and thinner materials. These results will be important in determining what additional modifications will be necessary to achieve the design goal for real time measurements to the skin and blood forming organs (BFO). Task 3: Modeling the Variance-Covariance Method The original proposal for the EVA dosimeter was based on the concept of having two independent proportional counters that would be used to obtain estimates of dose, D, and a quality factor, Q, based on estimating using the variance-covariance method. It was recognized that because of size limitations, the proportional counters would have to be located too close to one another to satisfy the condition that a single particle could not intercept both detectors. The additional constraint that one of the detectors must measure the dose at the skin surface and the other at a depth corresponding to the blood forming organs, makes the original variance-covariance method with paired detectors impractical. We have developed a method using a single detector in a variance-covariance scheme. The concepts are based on collecting the charge, q(i), in a single TEPC for n successive time intervals, i. The method proposed by Borak at Colorado State University (CSU) separates the data set into two groups of n/2 entries of values for q(i) based on odd and even indices. The n/2 pairs of data (odd and even) are used to obtain the covariance and each of the two sets of n/2 values (odd or even) to estimate a variance. Monte Carlo codes have been written to test the algorithmic using microdosimetric spectra obtained from measurements in Task 1. The tests indicated that if the change in dose rate between successive intervals was less than 1%, the single detector scheme provided reliable estimated dose rate and dose averaged lineal energy for estimating quality factors. The analysis also indicated that the estimate of dose mean lineal energy for high energy heavy ions (HZE) and recoil protons from PuBe neutrons converged to the correct value when the number of intervals exceeded 100 and the width of each interval was selected such that the mean number of events in each time interval (i) was less than 30.
More »This type of dosimeter has additional applications for first responders to nuclear accidents or terrorist events. It can also provide real time dosimetry during high altitude aviation and commercial space flight, diagnostic, and therapeutic medical procedures such as proton and carbon ion radiation therapy, and surveillance activities associated with homeland security and nuclear non proliferation. It can also serve an a area monitor with live-time capabilities that provide dose rate as well as estimates of quality factors for radiation workers as well as the general public.
More »Organizations Performing Work | Role | Type | Location |
---|---|---|---|
Johnson Space Center (JSC) | Lead Organization | NASA Center | Houston, TX |
This is a historic project that was completed before the creation of TechPort on October 1, 2012. Available data has been included. This record may contain less data than currently active projects.